Last Updated: 08/15/2019 in Frequently Asked Questions
The Patented MyoVision ScanVision 5 Electrode Adaptive Electrode Static sEMG was designed and patented with a provide the best skin contact possible through the use of a moving ground. This moving ground allows the electrodes to always make s...
Last Updated: 08/11/2019
How to locate Denial Responses for insurance denials.
Last Updated: 07/23/2019 in How do I...
How to adjust the amount of time the buttons need to be held in for to place a marker and how long they need to be held to start and advance DynaROM exams.
Last Updated: 07/23/2019 in Software Issues
This guide will help you troubleshoot, if it seems the MyoVision software is not registering your button presses.
Last Updated: 07/08/2019 in Hardware Issues
Learn how to troubleshoot when the "Bind" light on your SoloVision is blinking.
Last Updated: 07/02/2019 in Maintenance
Just like your car requires monthly maintenance, so do your ScanVision devices.
Last Updated: 07/01/2019 in DynaROM Questions
Question: What is muscle fibrillation? What causes muscle fibrillation? How does muscle fibrillation show up in exams? Answer: Muscle fibrillation is when the muscles fire at a high frequency and is a response to significant pain.This shows...
Last Updated: 07/01/2019 in Frequently Asked Questions
Do you know why you should be using QuickScan Pads? Read this article to find out!
Last Updated: 07/01/2019 in How do I... DynaROM
How to capture and edit screenshots of your tests to add to the Interpretation Report.
Last Updated: 07/01/2019 in How do I...
Summary This support article explains how to enter the calibration values for your Thermography device. Before you begin, make sure that your Thermography device has been setup in software settings. To do this, click on "Setup" at ...